Woodpecker Endo Gold Endomotor is mainly used in Endodontic treatment.During root canal preparation procedure,it is used to mold & clean the root canal.
a) Customized program sequence number 1-9, totally 9 programs
b) Operation mode
c) Battery consumption
d) Set speed
e) Set torque
f) File system
g) Operation mode
h) Torque setting
i) Real-time torque
Protective function of automatic reverse
During operation, if the load value exceeds the preset torque value, the file rotation mode will automatically change to Reverse Mode. And the file would return to normal rotation mode when the load is below the preset torque value again.
a) Clockwise rotation: Load value is lower than preset torque value
b) Counterclockwise rotation: Load value is higher than preset torque value.
c) Clockwise rotation: Load value is lower than preset torque value again.
- The device is suitable for shaping the root canals during root canal treatment & an advantage in replacing the hand files to rotary endodontics.
- Patients with implanted pacemakers (or other electrical equipment).
||Lithium battery in handpiece: 3.7V /1400mAh
|Input: ~100V-240V 50Hz/60Hz 0.4A Max
||0.6Ncm-5.0Ncm(6mNm ~ 50mNm)
||+5°C ~ +40°C
||30% ~ 75%
||70kPa ~ 106kPa
- This equipment should be stored in a room where the relative humidity is 10% ~ 93%, atmospheric pressure is 70kPa to106kPa, and the temperature is -20°C ~ +55°C.
- Avoid the storage in a too hot condition. High temperature will shorten the life of electronic components, damage battery, reshape or melt some plastic.
- Avoid the storage in a too cold condition. Otherwise, when the
temperature of the equipment increases to a normal level, there will be dew that will possibly damage PCB board.
Oiling of contra-angle:
Only the original oil injection nozzle can be used for oiling of contra-angle. After disinfection of contra-angle and before sterilization, oiling should be conducted under high pressure and high temperature.
- screw the injecting nozzle into jet of oil bottle. (Around 10
- Plug the nozzle into the end part of contra-angle, and then
grease the contra-angle for 2-3s till the oil flow out of contra-angle head
- Vertically place the end part of contra-angle or tilt the contra-angle to
let go the redundant oil under gravity.